The Newtonian telescope is very simple – Two mirrors, one curved and one flat. But to work, it needs to be set up properly with those mirrors and the eyepiece/focus tube aligned.
Hanging wallpaper in your house is easier when the walls, floors and windows are square!
Collimation is a very simple process made complicated by the number of variables involved and exacerbated by conflicting advice.
The aim is to get all the light that enters the aperture to fall on the primary (large) mirror and direct all of it to the secondary (small) mirror. Then all the light that is reflected off the secondary mirror should fall on the centre of the eyepiece.
Allen keys, Screwdriver – It is possible to replace the Allen bolts with wing nuts or knobs but I find using the key fine.
No tools Approximate alignment is possible by looking down the focus tube with no eyepiece. It helps to move your eye away from the opening.
Collimation Cap A simple cap (this can be a lens cap with a 1-2mm hole in the centre and a shiny washer glued to the inside) is essential can for accurately aligning the primary mirror. It also helps when aligning the secondary mirror.
Cheshire Eyepiece and Sight Tube Combination Tool A long tube with a side opening to let in light and a reticule (cross-hairs) is my “go-to” tool. The cross-hairs and sight tube makes it easier to confirm the secondary mirror is centred under the focus tube. What follows assumes that this is the tool being used but the information will be of use if a cap is used.
Collimated Laser Not essential in my opinion but as an additional check I use a collimated (this is important) laser to confirm that the secondary is facing the primary.
The views down the focus tube and the multiple reflections can be confusing. Here is what you see through the focus tube, sight tube and cap when the primary mirror has a centre ring mark
- Remove the aperture cover, tilt the OTA so that is almost horizontal and point it at a white surface like a white ceiling. A sheet of yellow paper is placed behind the focuser so the edge of the secondary can easily be seen.
- Remove all extenders. Insert the Cheshire tool using the 2″ adapter. Rotate the tool so the side aperture is facing a bright surface.
- I assume and occasionally check that the spider is centred and the focuser is at right angles to the OTA.
Move the focuser all the way in so that the mirror clips are as close to the edge of the secondary mirror as possible. Look through the small hole in the top of the Cheshire tool.
I want to check that the secondary is directly under the focuser. The tolerances for this step are larger than those for alignment of the primary mirror so I do not obsess about this check – good is good enough.
1. [Key starting point for placing the secondary mirror] The outer edges of the primary mirror and secondary mirror are circular (e=f) and concentric with the edge of the sight tube, so the secondary is centred (a=b and c=d) in respect of the focus tube.
2. The crosshairs of the sight tube intersect with the centre mark and the dark offset secondary reflection (x2) is offset toward the primary mirror. Therefore the primary mirror reflection is centred relative to the bottom edge of the sight tube.
This step can be confirmed with a collimated laser. I look through the aperture and check that the point of the laser sits in the centre mark of the primary. When the outgoing laser beam is aligned to the primary mirror centre mark (and the primary mirror is also aligned), the focal plane will be perpendicular to the eyepiece axis.
3. [This check is only valid if steps 1 and 2 have been confirmed] All three mirror clips are equally distant from the edge of the sight tube. As I withdraw the focuser, the clips disappear from view at the same time and the apparent size of the primary mirror increases relative to the secondary mirror. Therefore, all the primary mirror is presented to the focus tube.
Look through the Cheshire tool or Collimation Cap. The tolerances for the primary alignment are smaller than those for alignment of the secondary so I get this as accurate as I can.
- With a cap – The opening/pupil of the cap is centred in the primary centre mark/donut.
- With a Cheshire Sight Tube – The primary centre mark/donut is in the centre of the bright Cheshire ring.
- A star check will confirm that my primary mirror is aligned. Once my telescope has cooled down (this typically takes half an hour) and if seeing is good, I centre (this is important) Polaris in the eyepiece using high (200X) power. I move the focuser slightly in and then out and compare both views. The out of focus views show unbroken, concentric rings.
Correcting Secondary Errors
Good enough (having concentric circles) is fine for offset and rotation. The allowable secondary mirror tilt adjustment error tolerance (sight tube crosshairs relative to the primary mirror centre mark/donut) for a 200mm/8″ f/5.6 telescope is 6mm, and it’s easy to aim for an error of 3mm or less.
a ≠ b Offset (Axial, up-down) Error
If a > b then the secondary mirror needs to be moved away from the primary mirror
If a < b then the secondary mirror needs to be moved towards the primary mirror
Slightly loosen each of the three outer bolts with the Allen key. Grasp the secondary mirror in one hand without touching the reflective surface.
If a > b, loosen the outer bolts the same amount and tighten the centre screw with the screwdriver until a = b. You will need to keep loosening the outer bolts to allow the secondary mirror to move back. Do this gradually, loosening each of the outer bolts one turn.
If a < b, tighten each of the outer bolts (the same amount) to move the secondary mirror forward until a = b. You will need to tighten the centre screw with the screwdriver to “take up the slack”.
The above steps will introduce tilt error which will need to be corrected using the steps below.
c ≠ d / e ≠ f Rotation and/or Tilt Error
If each arm of the spider vane is equal and the focuser is level (see Initial Step above) then this is corrected by first rotating and then adjusting the tilt of the secondary mirror.
Correcting rotation is a coarse adjustment and correcting tilt is a fine adjustment.
[Once the a=b offset error has been corrected, then secondary errors will comprise the following rotation/tilt corrections.]
Rotation – Slightly loosen each of the three outer bolts with the Allen key. Grasp the secondary mirror in one hand without touching the reflective surface. Loosen the centre screw with the screwdriver and turn the mirror until the secondary mirror is circular and facing you. Tighten the centre screw carefully while holding the mirror.
This step will result in a considerable movement of the mirror. This adjustment is therefore coarse.
Tilt – Make small fine adjustments to each of the outer bolts until the sightube crosshairs are aligned to the donut/centre mark.
Recheck steps 1-3 above and repeat the above rotation/tilt corrections until the following reflections are as concentric as you can get them:
1) Bottom edge of the sight tube,
2) the actual edge of the secondary mirror and
3) the reflected edge of the primary mirror.
Ensure each outer bolt is reasonably tight.
Correcting Primary Errors
It is important to get this step as accurate as possible – for a 200mm/8″ f/5.6 telescope that is 1mm alignment error tolerance for the primary mirror but the Cheshire Sight Tube Combination tool magnifies the error by two, so you can easily improve the 1mm precision.
Carry out any corrections to the secondary mirror.
Tilt the OTA to approximately 45° and point it at a bright surface like a white wall.
Loosen the primary mirror locking bolts using the Allen key.
- Adjust one of the tilt screws on the primary mirror very slightly. Check for alignment (see above) to see if the centre mark/donut and centre of the bright Cheshire ring are getting closer or further away from each other.
- If they are getting closer, gradually turn the screw in the same direction. If they are getting further apart, turn the screw in the opposite direction. Stop turning when they stop getting closer or further apart.
- If the centre mark/donut and centre of the bright Cheshire ring are not aligned, move on to the next screw and repeat the above procedure.
- If necessary (it shouldn’t be) move onto the third screw and repeat the procedure.
When the primary mirror is aligned. Gradually tighten the locking bolts using the Allen key. Tighten each bolt gradually and in turn. Check alignment continually as it is likely that as you tighten the final bolt it will move the mirror. If this happens, loosen the bolt very slightly and very slightly loosen one or more of the other locking bolts. Repeat the tightening procedure, gradually tightening one bolt at a time until all three are tight.
This step requires moving from the mirror to the eyepiece and back again. It takes time if you are by yourself. However, once you have a feel for how the tilt adjustments affect the alignment you will find this process does not take long. I prefer to check alignment in daylight and my telescope only needs a primary alignment tweak every month or so. If the telescope has been transported in a car it is more likely to need aligning.
- Collimation is far easier to do than to explain – It is obvious when the mirrors are aligned as everything will be circular and (apart from the dark offset secondary reflection) centred.
- Collimation should never put anyone off buying a reflector telescope.
- The only aspect that may confuse you is the dark reflection of the secondary mirror. It is still circular but offset (not centred)
- My 8″ reflector has a solid tube and holds its alignment well. I check it once or twice a month and after I have transported the telescope.
- The error tolerance when making the above adjustments for a 8″/f5.6 telescope is between 6-8mm for the secondary and 1mm for the primary, but aim for the best you can.
- Collimation can become obsessive (and expensive) – Understanding tolerances helps keep OCD in check!
The study of Collimation can be fascinating and confusing at the same time. There are long detailed threads on astronomy forums that can contain conflicting advice. My advice is to find a guide that makes sense to you and has been written by an author you trust.
Acknowledgement: I am indebted to Vic Menard for his invaluable assistance in checking the accuracy of the above. I also wish to thank Jason Khaddar for his excellent diagrams and advice.